TOOLTOP TH202 4in1 256x192 Thermal Camera 2 Channels 80MHz Oscilloscope Multimeter PCB Thermal Imager Electrical Circuit Repair

2026-04-29T00:04:43-07:00

Price: 341.26

Note:

1. In order to correct non-uniformity, a mechanical control calibration plate is installed in front of the thermal imaging detector, which will be calibrated every 10 seconds to ensure more accurate measurements. This is a thermal imaging detector undergoing calibration work, which usually experiences a lag of about 1-2seconds. This is a normal phenomenon, so please don't worry.

2. Macro lens installed for PCB inspection at about 3 cm. For distances beyond this, use normal thermal imaging; otherwise the image will be blurred.

3. Mute setting: Select 【OSC】 or 【DMM】 mode, then press the 【MENU】 key, select Beep function and turn it off.

Features:

Professional 256x192 Thermal Imager

- 256x192 resolution 25Hz thermal imaging sensor

- With macro lens for 3cm close-range PCB inspection

- Real-time temperature profile recording improves inspection efficiency

2 channels Oscilloscope

80MHz analog bandwidth, 200MSa/s high-speed sampling, 12 customizable waveform parameters (such as time base), making circuit debugging more efficient.

20,000 Counts True-RMS Multimeter

Supports voltage/current/resistance/capacitance/frequency/duty cycle/diode/continuity and relative value measurements ,minimizes lead resistance and interference for more reliable results.

3.2 Inches Large Display

320x240 resolution, wide-view TFT color screen with 8 levels brightness adjustable. Readouts and waveforms are clear at a glance.

4GB Built-in Storage

Stores up to 2,000 data multimeter sets or 2,000 waveforms and supports saving thermal images and videos analysis.Facilitating experimental recording and data analysis.

3000mAh Lithium Battery

Rechargeable battery with a long battery life of approximately 5 hours, perfect for field measurements.

Specifications:

Thermal Imager Parameters

Sensor Type: VOx (Vanadium Oxide)

Thermal Resolution: 256x192

Image Capture Rate:25Hz

Temperature Range: -20°C~+550°C

Measurement Accuracy:±2°C or ±2%

Emissivity: 0.01~0.99

Palette: White Hot, Black Hot, Fusion1, Fusion 2, lron 1, lron 2,Deep Blue, Color 1, Color 2.Lava, Rain, Green, Red Hot.Deep Purple

Field of View(FOV):50.0°(H)x37.2°(V)

Focus:3.5mm (fixed, non-adjustable)

Measurement Points: 5(center,highest,lowest,area max, area min)

lmage Format:JPG (photo), AVl (video)

Storage Capacity:TF card

Oscilloscope Parameters

Analog Bandwidth: 80MHz(Single Channel),40MHz(Dual Channel)

Maximum Sampling Rate:200MSa/s

Channels:2

Storage Depth: 2K

Vertical Resolution: 8 bits

Coupling Mode:DC、AC

Input lmpedance: Approx.1MΩ

Trigger Mode:Auto、Normal、 Single

Trigger Edge Selection:Rising / Falling edge

Vertical Sensitivity Range: 10mV/div-10V/div

Vertical Accuracy: ±(5%+ 0.2div)

Voltage Measurement Parameters:Vp-p、Vp、Vmax、Vmin.Vavg、Vrms

Time Measurement Parameters:Freg、Period、 +Width、-Width、+Duty、-Duty

Non-Linearity: ±1 bit

Attenuation:Vertical ±4.0, Horizontal ±6.0

Time Base Range:6ns/div-50s/div. adjustable in 1-2-5 steps

Horizontal Accuracy: ±(0.01% + 0.1div)

Automatic Measurement Accuracy: ±(5%+ 0.2div)

Multimeter Parameters

Display: 20000 Counts

DMM Measurement Speed: 3 times/second

DC Voltage: 200.00mV/2.0000V/20.000V/200.00V/1000.0V

AC Voltage(True RMS):200.00mV/2.0000V/20.000V/200.00V/750.0V

DC Current: 2000.0mA/10.000A

AC Current(TrueRMS):2000.0mA/10.000A

Resistance:200.00Ω/2.0000kΩ/20.000kΩ/200.00kΩ/2.0000MΩ/20.000MΩ/200.00MΩ

Capacitance:9.999nF/99.99nF/999.9nF/9.999uF/99.99uF/999.9uF/9.999mF/99.99mF

Frequency:9.999Hz-9.9999MHz (input ≥ AC 500mVp-p; higher frequency requireshigher input voltage)

Diode Test:Open-circuit voltage approx.3V, maximum test current approx.2mA

Continuity Test: Buzzer sounds at<50Ω; displays OL at ≥200Ω

Auto Range Switching: Applicable to all measurement functions

Overrange Indication： O.L(Over load)

Dynamic Data Window: Continuously refreshes to display real-timemeasurement data

Fuse :10A/250V

Signal Source Parameters

Waveform Frequency

Sine Wave:10Hz-100kHz(1-2-3-5 step mode)

Triangle Wave:10Hz-100kHz(1-2-3-5 step mode)

Square Wave:10Hz-4MHz (5MHz for reference only)(1-2-3-5 step mode)

Square Wave Duty Cycle:10%-90%(adjustable in 10% increments, 1-2-3-5 step mode)

Output Voltage: 3Vpp (output impedance approx.50Ω)

Other Parameters

Display: 320x240; 3.2-inch TFT color screen

Backlight Brightness: 1-8 levels adjustable

Battery: 3000mAh rechargeable lithium battery

Low Battery Indicator: Support

Operating Time: About 5 hours

Operating Environment: 0℃~+40℃; <75%RH

Overall Dimensions: 173mmx89mmx38mm

Silent Mode: Supported

Auto Power-off: 5min-60min or disabled

Power : DC5V/2A ( charger not Included)

Storage Capacity: DMM: 2000 records; OSC: 2000 waveforms

Storage Environment: -10℃~+60℃; <90%RH

Weight: 296g/10.4 0z (0.65 lb)(excluding accessories)

Package List:

1 x Thermal Imager Oscilloscope

2 x Oscilloscope Probes

1 x One Pair Test Leads

1 x BNC to Alligator Clip Cable

1 x USB Cable

1 x EVA Bag

1 x English User Manual